Surgical shadowless lights can provide a suitable lighting environment and effectively reduce visual fatigue, and, in the process of using surgical shadowless lights, different types of surgical shadowless lights and different methods of use can also lead to the use of surgical shadowless lights. The following, the manufacturer of the surgical shadowless lamp is asked to give you a detailed introduction to this part of the content.
1. The control box is placed in the wall, so the heat generated by its transformer cannot be dissipated, which is easy to cause the transformer, brightness adjustment board, transformer output line burned. If the control box is drilled with a few holes, using compressed cotton to block the hole, so that it is conducive to heat dissipation and can well filter dust.
2. The surgical shadowless lamp head back cover is heavy, hanging in mid-air can easily turn, but disassembly is not convenient. In order to strive for operating time, if the main body of the surgical shadowless lamp back cover applies the button structure, it can be easily opened and closed, which is conducive to emergency troubleshooting.
3. The back cover of the main body of the surgical shadowless lamp is sealed tightly, the heat generated by the bulb can not be distributed, resulting in the back cover of the main body of the shadowless lamp has been burned off many times. If a few holes are punched in the control box, blocked with the compressed cotton, the phenomenon of line burning on the back cover of the main body of the shadowless lamp will be reduced.