The manual hydraulic transmission operating table uses a cylinder as the actuator to perform the individual movements of the console surface. Most of the posture movements are achieved by hand-controlled pneumatic spring boosting system, and the lifting action is completed by the foot control oil pump system driving the column device.
Electric hydraulic operating table is powered by electro-hydraulics, according to the clinical needs of the operating bed, press the corresponding button of the controller, through the control motherboard, start the hydraulic motor, the hydraulic pump can work, so as to control the operating bed for various position conversion, such as lifting, left and right tilt, mobile fixation and other functions, so that it can meet the requirements of surgical operation. Releasing the key, the table movement stops immediately and the operating table stays in that position. The operating bed is equipped with a limit switch, which is used to limit the operation of the operating bed to always operate within a safe range.